Instruction Manual. Product Overview. Contents of this Manual. Chapter How to Use the Communicator. How to Use the DataViewer.
|Published (Last):||7 January 2018|
|PDF File Size:||15.91 Mb|
|ePub File Size:||11.74 Mb|
|Price:||Free* [*Free Regsitration Required]|
Manual SSJ Engineering Geology Ma DVR watch manual V1. Conditions for use This software may not be bundled or packaged with any other software or other publications which are sold for profit without the express written permission of the copyright owner. Further, this product may not be reverse-engineered, decompiled, or disassembled without the express written permission of the copyright owner.
Responsibility for use This program is provided as freeware for use at the user's risk. Corporation disclaims any and all responsibility for any consequences arising out of use of this software, as well as for any programming errors or problems arising out of use of the software.
In the following example, 2 units of Model are currently connected to the PC. The assigned name will appear in the Connection List in place of the serial number of the Operate or manage the by clicking the icons on the left column. Operate the using these icons. Information about the selected is displayed here. Serial number of the selected Up to 12 characters can be used. Even after turning off the power to the , the assigned name will not be deleted.
Displays the number of saved BMP images in the selected Displays the number of measurement data saved in the selected Downloading Data from Model All screen shot images saved in the will be automatically saved in a new folder under the designated directory on your PC.
All measurement results saved in the will be saved in a new folder under the designated directory on your PC. Example: Saving measurement data on February 15, Internal memory not A folder named B will be created and partitioned and 1 set of Ex.
Clicking this button will launch the DataViewer application. You can also choose to open the same application using the Start menu on your PC. Regardless of the operating state of the PQA whether on standby or recording data can be downloaded at will. Choose how to display the screen of the as well as perform other display functions. Remote Display The screen of the selected from the Connection List will be displayed. Remote Operation Keys Click the virtual keys with your mouse to remotely control the selected as you would on the device.
A window similar to the one shown below will appear. Choose the folder containing the data you wish you view e. All available measurement data folders will appear next to an image icon of the Settings The event settings will be displayed as shown below. Measurement Settings Information Measurement settings and system settings are displayed as shown below. Event List This window displays the event list.
The contents in the event list are the same as the event list on Model The event to view or analyze can be selected by clicking on the item on the event list. Also, up and down cursor keys can be used to select the event. Channel 3 Displayed for voltage swell, voltage Event in dip, interruption or transient Event out overvoltage.
When both an event voltage fluctuation graph and inrush current graph exist for a single event, the inrush current graph icon will be displayed. Event Details Displays the event details of the selected event in the event list. The contents of the event details are the same as those appearing in the event details screen on Model Displayed for voltage swell, voltage dip, Minute:Second interruption, and Displayed to milli-second units transient over voltage.
Maximum voltage for the voltage swell Displayed for the event out of the voltage swell, Residual voltage for the voltage dip voltage dip and Residual voltage for the interruption interruption. You can also scroll in the horizontal axis direction. Change the current axis magnification by selecting in x1, x2, x5, x10, x20 or x The magnification rate of the vertical axis can be changed and scrolled. Scroll bar??
Voltage RMS of U1??? Voltage RMS of U2?? Voltage RMS of U3 Change the vertical axis magnification rate by selecting in x1, x2, x5, x10, x25 or x Current RMS of U1?? Current RMS of U2?? Current RMS of U3 Change the vertical axis magnification by selecting in x1, x2, x5, x10, x25 or x The magnification rate of vertical axis can be changed and scrolled.
Scroll bar??? Maximum vale in an interval Average value in an interval Minimum value in an interval Select the item to display using the pull down menu. Change the vertical axis magnification rate by selecting from x1, x2, x5, x10, x25 or x Displays the recording interval setting on Model Displays the number of data recorded by Model Click this button to save the measurement data in CSV format. U1 is outside the set nominal voltage range. Measurement value is outside the set frequency range.
Function Specifications of Communicator Remote Operation Function Functions: Display the same screen as the display on Model with automatic refresh. Operate Model using the virtual keys. Function Specifications of Data Viewer All reasonable care has been taken in the production of this manual, but if you find any points which are unclear or in error, please contact your supplier or the International Sales and Marketing Department at HIOKI headquarters.
In the interests of product development, the contents of this manual are subject to revision without prior notice. Unauthorized reproduction or copying of this manual is prohibited.
Hioki 3197 Measurement Manual
3197appli Manual e 01
Providing the finest test equipment solutions since 1992
POWER QUALITY ANALYZER 3197 Application(64-bit)